How does dental curing light work
The curing light is an oral device for repairing teeth. It uses the principle of light curing to quickly cure the dental repair resin material under the action of light waves within a specific wavelength range, thereby filling the tooth cavity or bonding the bracket.
The first is the strong light mode: the strong light indicator light is on, press the power switch to enter the strong curing light dental mode,
The buzzer will make a “beep” sound every 5 seconds, and the running time will end automatically after the countdown is completed.
The second is the pulse mode: the pulse indicator light is on, press the power switch, the power indicator light is on, you can enter
In pulse curing mode, the buzzer will send out a “beep” sound every 5 seconds, and automatically after the countdown of the running time is completed
The third is the gradual bright mode: the gradual bright indicator light is on, press the power switch, the power indicator light is on, you can enter
Gradual bright curing mode, the buzzer will sound “beep” every 5 seconds, the light source will gradually brighten from weak light to strong light
Shooting, the running time will end automatically after the countdown is completed.
The dental curing light can customize the working time of the setting mode, up to 40 seconds, long press MENU to enter the setting state
state, short press to select mode and long press to exit setting, press POWER key to select the corresponding time.
This uv curing light has an intelligent battery detection function, which will alarm and automatically shut down when the battery power is low.
To charge in time. .
The device automatically shuts down when there is no key operation after one minute, the power indicator and function indicator
off. To turn it on, press the power switch again to turn it on.
The charging method of the rechargeable LED curing light machine is constant voltage charging, and the charging time is about 8 hours.
Three working mode options
Large battery capacity: Li-ion battery 3.7V, 2200mA/h;
Optical power: greater than 1600mw/c㎡;
Long standby time: more than 90 days Single weight: 0.4kg